TBQ-2C is Mainly used to measure the wavelength of 0.3-3.2 micron solar total radiation. The meter is composed of double layer quartz glass cover, induction element, light screen, desiccant and other parts.
The induction element, which is the core of the table, consists of a fast responding wire-wound electroplating thermopile. The induction surface is coated with 3M matt black paint. The induction surface is a hot contact. When the temperature rises when there is sunlight, it forms a thermoelectromotive force with the cold contact on the other side, which is proportional to the solar radiation intensity.
The double glazing is designed to reduce the effect of air convection on the radiation table. The inner cover is designed to block the infrared radiation of the outer cover itself.
|Measuring angle||2π solid angle, 0-90°|
|Zero drift (temperature drift:5k/h)||±5W/m2|
|Cosine correction||≤±7%(Solar elevation angle=10°)|
|Recalibration interval||2 years|
|Desiccant||Silica gel desiccant|
|Pack||Aluminum alloy instrument box|
|Installation bracket(optional)||Horizontal bracket or adjustable angle bracket|